Contact UsTwo Pershing Square 2300 Main Street, Suite 170 Kansas City MO 64108
Two Pershing Square
2300 Main Street, Suite 170
Kansas City, MO 64108
Telephone: (816) 221-6600
Toll Free: 1 (877) 284-6600
Fax: (816) 221-6612
FDA Letter to Health Care Providers
Posted by the FDA on April 25, 2018.
The Issue: FDA is providing preliminary information concerning magnetic resonance (MR) thermometry reliability with magnetic resonance-guided laser interstitial thermal therapy (MRgLITT) devices.
The FDA is currently evaluating data which suggests that potentially inaccurate MR thermometry information can be displayed during treatment. For example, MR parameters such as voxel size (measurement of the image resolution or detail) and MR image acquisition time (e.g., up to 8 seconds) may contribute to inaccurate MR thermometry readings and potential errors in the ablation assessment. In addition, MRgLITT devices may not account for the continued thermal spread of energy to the surrounding tissue (as the target ablation area returns to its baseline temperature), which may result in an underestimation of thermal damage.
The FDA has reviewed Medical Device Reports (MDRs) and literature reports which describe adverse events such as neurological deficits (e.g., focal motor deficits, aphasia, cognitive changes), increased intracerebral edema or pressure, intracranial bleeding, and/or visual changes (e.g., visual field deficits, blurry vision) when these devices were used to treat intra-cranial lesions. Several of these reports note events which required urgent medical and/or surgical intervention, and may have been associated with patient deaths. However, it is unclear at this time, whether an inaccuracy of MR thermometry directly caused or contributed to these events.
MRgLITT devices are commonly used in neurosurgical procedures for minimally invasive ablation of brain tumors, epileptic foci, or radiation necrosis. During these procedures, the tip of a laser probe is stereotaxically placed within the tissue to be ablated, and controlled thermal energy is delivered to the target tissue. MR thermometry is used to monitor the changes in temperature at the treatment site using a Magnetic Resonance Imaging (MRI) scanner, and proprietary software information is displayed on a workstation to aid the physician in assessing and controlling thermal damage.
If you would like to receive news and blog updates on a regular basis, sign up to receive our email newsletter. Your email address will only be used to send you our newsletter and respond to inquiries.
Past results afford no guarantee of future results and each case is different and is judged on its own merits. The choice of a lawyer is an important decision and should not be based solely upon advertisements.